Store up to 50 measurements to compare body temperature changes
Parameters setting is adjustable
Keep away from fire.
Keep away from child.
See doctor if necessary.
Passive infrared detector
Open battery cover, install two AAA batteries. Make sure battery
1. Press ON button,LCD display all readings, it means that it is in self-check mode for 2 sec.
2. “ Beep”s o und w ill b e h ea rd a f ter self-check, start measuring and LCD displays reading, Backlight green, yellow, red (one of three colors) will be lighten according to measuring readings.
Low fever setting: Yellow LCD backlight
High fever setting: Red LCD backlight with alarm
Hold on for at least 30 sec after 5 times repeat measurements.
Turn on thermometer，quick press (MODE）button to switch mode from“BODY ”to“OBJECT”（Default setting is“BODY”mode.)
Memory data checking
Quick press (MEMO) button to check previous memory data.
1. The more MEMORY log number, the earlier memory data, the less MEMORY log number, the later memory data.
2. If no memory data, then shows“－－－－”. 03.The thermometer can store up to 50 sets measuring data, over 50 sets, it will be covered by latest one.
Clear memory data
Turn on the thermometer, long press MEMO button till LCD displays "clr”，and a long “beep”indicates all data cleared.
Default Setting Change
Default settings can be adjusted as follows:
【Switch ℃ or ℉ 】 F1（Default setting is ℃), Turn on thermometer,long press SET button to enter into F1，release button goes into F1 parameters，press MEMO or MODE,button to switch between Celsius ( ° C) and Fahrenheit( ° F). Confirm setting by press SET button, and enters into F2.
【High fever alarm setting 】F2（Default setting is 38.1℃）Enters into F2, release button goes into F2 parameters, press MEMO button once to increase 0.1℃, press MODE button once to reduce 0.1℃, long press MEMO and MODE button to increase or reduce more ℃, confirm setting by press SET button and enters into F3.
【Alarm ON/OFF setting】F3（Default setting is alarm on）Enters into F3, release button goes into F3 parameters, press MODE or MEMO button to ON/OFF alarm, confirm setting by press SET button and enters into F4.
【Temperature deviation setting 】F4（ Default setting is 0）It is necessary to calibrate the thermometer when environmental temperature deviates.Enters into F4, release button goes into F4 parameters, press MEMO button once to increase 0.1℃, press MODE button once to reduce 0.1℃, long press MEMO and MODE button to increase or reduce more ℃, confirm settings by press SET button and exit setting mode. F4 parameter adjusting range is -5℃~ 5℃。